Principles and Equipment of Dermoscopy
Equipment, Principles, Non polarized dermoscopes, Non polarized dermoscopy, Polarized dermoscopy, Surface glare 01-Principles and Equipment – cite! 01-Principles and Equipment (message) 01-Principles and Equipment – participate!
Principles of dermoscopy
The clinical "naked eye" examination is important in the assessment of the gross morphological features of the lesion, such as size, shape, colors, contours, and surface topography.
Since the refractive index of the stratum corneum (surface layer of the skin) is higher than that of air, much of the incident light is reflected off the surface of the skin and causes the surface glare, as the reflected light overwhelms the retina and precludes the observer from visualizing the light reflected from the deeper layers of the skin. Thus, clinical "naked eye" examination mainly allows the assessment of morphological features of the surface layer of the skin (stratum corneum), and to a much lesser extent, the colors and structures of the deeper layers of the epidermis and the superficial dermis.
In body areas with a thick stratum corneum (e.g. on acral surfaces), there is a greater back-scatter of the light and the skin appears more opaque. In body areas with a thin stratum corneum (e.g., eyelids), the skin appear more translucent.
A dermoscope (or dermatoscope) is a handheld device, equipped with a magnification lens and a light source. It enables the visualization of the subsurface morphology of cutaneous lesions, down to the depth of the superficial dermis.It reveals colors and structures that are normally not visible to the unaided eye and improves the diagnostic accuracy and confidence level of experienced users, for both pigmented and non-pigmented skin lesions.
Non-polarized dermoscopes (NPD) are equipped with a magnification lens and light-emitting diodes to provide illumination. They require direct contact of the glass plate with the skin surface, and the presence of a liquid interface with a refractive index equal to or closely matching that of the skin. Different immersion liquids can be used: 70% alcohol, gel (i.e., ultrasound gel, antibacterial gel), water or mineral oil. Air bubbles between the dermoscope’s glass plate, the immersion liquid, and the skin surface create a skin–air interface. The interface causes back-scatter of light and precludes the observer from visualizing structures below the stratum corneum. Thus, air bubbles should be eliminated.
NPD allow visualization of subsurface structures located in the epidermis and the dermal-epidermal junction (DEJ), but can hardly visualize structures deeper than the DEJ.
Optical properties of light during the use of NPD
- The Surface glare is eliminated by matching the refractive indexes of the NPD's glass plate, the immersion medium, and the skin.
- Superficial penetrating light is the main source of contrast when using NPD. The light that enters the skin is absorbed (e.g. by melanin) or reflected back (e.g. by keratin in milia-like cysts) at the layers of the epidermis and the DEJ. The superficial penetrating light undergoes minimal scattering events and is therefore the main source of light that reflects back to the NPD lens.
- Deep penetrating light contributes only a small fraction of back-reflected light detected with NPD. This is due to the decay of light by multiple scattering events, as it goes deeper into the skin.
Melanoma, as it appears clinically (insert) and on NPD:
Globular nevus as it appears clinically (insert) and on NPD:
Polarized dermoscopes (PD), like non-polarized dermoscopes, contain light-emitting diodes to provide illumination and are equipped with a magnification lens. However, PDs use two polarized filters to achieve cross-polarization (see below). Hence, they do not require direct contact with the skin, and do not require the use of immersion liquids.
PD allow visualization of subsurface structures located at the dermal-epidermal junction (DEJ) or superficial dermis, and they are nearly "blind" to the skin's surface and to structures in the superficial epidermis (e.g., comedo-like openings).
Some PD devices allow the user to opt between non-contact PD and contact PD. Under contact PD, the use of an immersion fluid (e.g. 70% alcohol) can enhance image quality, probably by allowing more source light to enter through the stratum corneum.
Optical properties of light during the use of PD
- Light emitted from the dermoscopy unit (source) passes through a polarizer, resulting in the generation of polarized (unidirectional) light.
- Light reflecting back toward our eye (detector) must first pass through a cross-polarized filter whose direction is perpendicular (orthogonal) to that of the source polarizer.
- Polarized light cannot pass through the cross-polarizing filter unless the light changes its direction by 90°, which occurs if the original polarized light undergoes sufficient scattering events in the skin that change its direction ("randomization of polarization").
- Surface glare: maintains its original polarization, and thus cannot pass through the cross-polarized filter. Therefore, PD is "blind" to surface glare.
- Superficial light: does not undergo enough scattering events to result in randomization of polarization. Therefore, PD is also "blind" to back-reflected light from the superficial layers of the epidermis.
- Deep penetrating light: reaches the level of the DEJ and superficial dermis, undergoing multiple scattering events that result in randomization of polarization. Thus, back-reflected "deep penetrating light" can pass through the cross-polarization filter and enter our eye, allowing the visualization of dermoscopic structures from the DEJ and superficial dermis.
Shiny white streaks, a feature only visible on PD:
Differences between polarized and non polarized dermoscopy
The main difference between non-polarized dermoscopy (NPD) and polarized dermoscopy (PD) is the depth of visualized structures. While NPD is better for inspecting structures in the superficial skin layers (e.g., superficial epidermis down to the dermo-epidermal junction [DEJ]), PD is better for evaluating the deeper skin layers (e.g., DEJ and superficial dermis).
For example, both Milia-like cysts and blue-white veil are caused by superficial changes in the epidermis and are therefore better visualized with NPD. On the other hand, Shiny white structures (chrysalis/crystalline, blotches and strands, rosettes) are better visualized with PD since they are associated with increased collagen at the superficial dermis. In addition, polarized light rapidly randomizes its polarization when it encounters a birefringent structure, such as collagen.
An example of Milia-like cysts,better visualized under NPD:
An example of a blue-white veil, also much better visualized with NPD:
And finally an example of Shiny white lines/streaks, which are much better visualized with PD:
Additional difference between NPD and PD
- Since PD does not require direct skin contact, blood vessels and pink color (vascular blush) are more evident under PD (due to lack of pressure effect). In addition, blood vessels are located in the dermis that is better visualized with PD.
- In pigmented lesions with melanin at the DEJ (e.g.. junctional nevi) or superficial dermis (e.g. blue nevi), PD will show slightly darker shades of brown and blue, and sometimes more variability in pigmentation, compared with NPD.
The differences between PD and NPD may impact the diagnostic accuracy and diagnostic confidence level. For example, PD can increase sensitivity for detecting amelanotic melanomas or structure-poor melanomas and basal cell carcinomas, because PD highlights the presence of blood vessels, vascular blush, and/or white shiny lines (chrysalis/crystalline). In contrast, NPD can increase specificity by allowing to correctly identify Milia-like cysts and comedo like openings in seborrheic keratoses.
”Hybrid” dermoscopes toggle between PD and NPD modes. These hybrid devices can enhance dermoscopic diagnosis because PD and NPD provide complementary information. These devices should always be in direct contact with the skin and used with a liquid interface, otherwise dermoscopic structures will only be visualized in PD mode.
|Colors and structures||NPD||PD|
|Blue-white due to orthokeratosis||+++||+|
|Blue-white due to regression||+++||++|
|Chrysalis or white scar||+/−||+++|
Equipment for Dermoscopy
There many different types of handheld scopes. There are scopes that use polarized light or standard LED light. There are scopes that allow to toggle from polarized to non-polarized mode. Some scopes can also be attached to a camera (smartphone / tablet).
It is the author's preference to use scopes that allow toggling between polarized and non polarized light because it provides additional information. When asked which scope is most recommended, the author's answer is - Try it out, hold it in your hand and see which ones feels best for you.
Here are some examples of handheld dermatoscopes that are currently on the market:
Systems for digital dermoscopy
There are many different systems for digital dermoscopy on the market. Every system has different features, including Storage and retrieval of lesions (clinical and dermoscopy images); total body imaging; and digital mole monitoring.
In addition, some systems use computer assisted features:
- Computer assisted total body imaging
- Computer assisted diagnosis of dermoscopy images
- Computer assisted hair and scalp analysis
- Automated PASI calculation
- 3D imaging
Every system is very different than the other, and the best thing is to try them out in real world conditions in order to find out if the workflow of the software is compatible with your individual workflow. Some examples include:
Cases / Attachments for smartphones
Several companies provide Dermoscope cases adapt to a smartphone or a tablet that provide good image quality. The images can be stored on the phone in specific apps and then be transferred to the medical record or systems for digital dermoscopy.
Attachments (lenses) for digital cameras
Dermoscopic lenses that attach to digital cameras provide very high quality images. Examples include:
Full body imaging systems
Apart from classic digital dermoscopic systems, there are vendors enabling the users to aquire and organise full body images in 2D or 3D. They are either integrated into digital dermoscopic systems, work with smartphones, or have a separate rigging system. Examples include:
Please contact us if your device is not listed in this chapter by sending an e-mail to
This chapter is based on the previous basal cell carcinoma chapter which can be found here: https://dermoscopedia.org/Dermoscopic_equipment
- An Atlas of Dermoscopy, Second Edition. Marghoob A. et al. CRC Press; 2012.
- Agero, A.L., Taliercio, S., Dusza, S.W., Salaro, C., Chu, P. & Marghoob, A.A., 2006, Conventional and polarized dermoscopy features of dermatofibroma. Arch Dermatol, 142, 1431–7.
- Anderson, R.R. & Parrish, J.A., 1981, The optics of human skin. J Invest Dermatol, 77, 13–19.
- Bafounta, M.L., Beauchet, A., Aegerter, P. & Saiag, P., 2001, Is dermoscopy (epilumi- nescence microscopy) useful for the diagnosis of melanoma? Results of a meta- analysis using techniques adapted to the evaluation of diagnostic tests. Arch Dermatol, 137, 1343–50.
- Benvenuto-Andrade, C., Dusza, S.W., Agero, A.L., Scope, A., Rajadhyaksha, M., Halpern, A.C. & Marghoob, A.A., 2007, Differences between polarized light der- moscopy and immersion contact dermoscopy for the evaluation of skin lesions. Arch Dermatol, 143, 329–38.
- Benvenuto-Andrade, C., Dusza, S.W., Hay, J.L., Agero, A.L., Halpern, A.C., Kopf, A.W. & Marghoob, A.A., 2006, Level of confidence in diagnosis: clinical examination versus dermoscopy examination. Dermatol Surg, 32, 738–44.
- Gewirtzman, A.J., Saurat, J.H. & Braun, R.P., 2003, An evaluation of dermoscopy fluids and application techniques. Br J Dermatol, 149, 59–63.
- Kelly, S.C. & Purcell, S.M., 2006, Prevention of nosocomial infection during dermoscopy? Dermatol Surg, 32, 552–5.
- Kittler, H., Pehamberger, H., Wolff, K. & Binder, M., 2002, Diagnostic accuracy of dermoscopy. Lancet Oncol, 3, 159–65.
- MacKie, R.M., 1971, An aid to the preoperative assessment of pigmented lesions of the skin. Br J Dermatol, 85, 232–8.
- MacKie, R.M., 1972, Cutaneous microscopy in vivo as an aid to preoperative assessment of pigmented lesions of the skin. Br J Plast Surg, 25, 123–9.
- Marghoob, A.A., Cowell, L., Kopf, A.W. & Scope, A., 2009, Observation of chrysalis structures with polarized dermoscopy. Arch Dermatol, 145, 618.
- Pan, Y., Gareau, D.S., Scope, A., Rajadhyaksha, M., Mullani, N.A. & Marghoob, A.A., 2008, Polarized and nonpolarized dermoscopy: the explanation for the observed differences. Arch Dermatol, 144, 828–9.
- Ronger, S., Touzet, S., Ligeron, C., Balme, B., Viallard, A.M., Barrut, D., Colin, C. & Thomas, L., 2002, Dermoscopic examination of nail pigmentation. Arch Derma- tol, 138, 1327–33.
- Stauffer, F., Kittler, H., Forstinger, C. & Binder, M., 2001, The dermatoscope: a potential source of nosocomial infection? Melanoma Res, 11, 153–6.
- Wang, S.Q., Dusza, S.W., Scope, A., Braun, R.P., Kopf, A.W. & Marghoob, A.A., 2008, Differences in dermoscopic images from nonpolarized dermoscope and polarized dermoscope influence the diagnostic accuracy and confidence level: a pilot study. Dermatol Surg, 34, 1389–95.